Xitron XT2640 Precision Multi-Channel Power Analyzer
The XiTRON XT2640 Series Precision Multi-Channel Power Analyzers is actually three power analyzers in a single chassis with a single user interface. These are called Virtual Power Analyzers (VPA) or Channels. The XT2640 may have up to 4 channels installed, which may be any combination of channel cards and with any combination of available current input options.
Channels may be configured in any one (or none) of the 3 virtual power analyzers. Each virtual power analyzer may be configured for up to all channels installed.
Each VPA is independently configured for multi-channel wiring configuration, signal filtering, default measurement coupling, display results smoothing and significant digits, VA/VAR combine method, and efficiency grouping. VPAs may optionally be configured to be synchronized to each other.
XiTRON Technologies, founded in 1990, is the premier source of precision power testing and measuring equipment for industrial and consumer product development and manufacturing. XiTRON’s sophisticated technology provides companies the edge in design verification and product manufacturability. XiTRON is ISO9001- 2008 certified.
After the XT2640 has been shipped or otherwise handled in an unknown manner, you should visually inspect the XT2640 for damage before attempting to operate it. Particular attention should be taken to ensure that there are no significant dents or cracks in any outer surfaces, there are no marks or scratches on the front panel LCD touchscreen, and that all terminals are securely mounted to the unit and are not cracked or significantly dirty. If any abnormality is noted then it is recommended that the XT2640 be serviced prior to being placed into use, as safety may have been compromised.
- AC to DC Power Supplies (single or multiple outputs)
- DC to DC Power Supplies (single or multiple outputs)
- DC or fixed frequency AC supplied variable or fixed speed motor drives
- Power transformers (single or multi-phase)
- Lighting ballasts (most types)
- Standby or backup power supplies (AC or DC)
- Photovoltaic power generators (DC-in; DC- or AC-out)
- Mechanical motor/generator applications
POWER DATA Screen
This displays V, A, W, VA, VAR and PF data for any selected channel or the totals for any selected analyzer. Along with the primary selected data the user may select to display further details such as non-default coupled data and peak data. Screens are provided for displaying loading and integrated data results. POWER DATA Screen enables you to view numerical results for all voltage, current, and power (W, VA, VAR and PF) measurements in peak, RMS, fundamental only or rectified forms, inrush, resistive/inductive/capacitive loading, integrated and long-term average measurements such as Watt-Hours, efficiency and power loss.
CUSTOM DATA Screen
The user may build a custom data screen showing any of the available results from any of the channels and/or analyzers using a spreadsheet type format (one heading cell, and a 4×14 array of cells). Each cell may be defined as showing a measurement result (with or without preceding text and with or without units shown), or showing text or blank and each cell may be set for a selection of font sizes and colors. The definition for this screen may be constructed on an external computer using a free application provided with theXT2640 Series and then downloaded directly into the XT2640 using an interface or imported from a drive connected to the front panel USB port of the XT2640 (in this later case the number of available custom screens is only limited by the drive).
As previously described, the XT2640 Series continuously maintain an historical record of most measurement results (all non-harmonic results and selected harmonic results). The recorded values for up to 4 of these can be graphically displayed using the HISTORY screen. The user selects the data record to show, whether to display the trend, trend and extents, or the extents of the data within each data record, and also set the scaling, centre offset and color of the plot for each trace. The user may display the historical data for the entire recorded period of time. In this case the plot automatically scales horizontally to provide the entire span of time (from 5 seconds to 400 days total span) and includes a cursor which may be moved throughout this period by the touch screen (the cursor is always at a fixed time, not a fixed position in the screen). The actual values at the cursor position are listed numerically. The user may also display the historical data for a ‘zoomed’ time interval. In this case the user may select to zoom to a period surrounding the cursor position on the full history screen and then zoom in/out to a span showing ¼, 1/16th or 1/64th of the total time of the present history record and may move the zoomed display left or right using the touch screen. The cursor is always in the center of the displayed timespan.
Cycle View represents a single cycle of the voltage and or current periodic waveforms. The waveforms in photo section represent a full 10V square wave in blue and a 50:1 zoomed in view in red. Since the user sets amplitude and scaling – the result is an almost unlimited ability to amplitude zoom to expose fine detail. The sampling is forced to be asynchronous to higher order harmonics which leads to an effective sampling rate of 384MSPS.
Scope View offers waveform acquisition and analysis similar to a digital scope. Up to six signals can be displayed, each havinguser selectable scaling, offset and color. Time-base, trigger and trigger position are all user selectable. However, with amplitude accuracies as high as 0.03% – you are unlikely to find any other scope with this high level of precision.
This displays a bar chart showing all available harmonics for any selected signal(V or A) and also a listing of any 8 harmonics numerical amplitudes and phases (scrollable within the available harmonics). The bar chart may be configured to show linear,relative linear, logarithmic or relative logarithmic amplitudes. The user may also import (via an interface or from a drive via the front panel USB port) a set of V and/or A harmonic limits which may optionally be applied. If a set of limits is available and has been enabled for the selected signal, the bar chart displays a limit line and also displays any failing harmonics as a red color above the limit.The XT2640 Series have the ability to compare voltage and current harmonicsagainst limits set for each harmonic independently for voltage and current.
This displays a polar chart graphically showing the fundamental voltage and current vectors for a selected channel or analyzer. As applicable for an analyzer theinter-phase voltages and non-measured phase (e.g. neutral) vectors are also displayed.The user may individually enable/disable the display of each vector (up to 10 can be shown at any given time) and also set the color of the vector. Additionally, thenumerical amplitude and phase of each vector is shown as a list and (for 3ø4w configuration) the positive, negative and zero sequence amplitudes are listed.
Fundamental Vector Analysis is achieved as the fundamental voltage and current vectors for each virtual power analyzer and for each channel canbe viewed graphically on the screen and the numerical values(amplitude and phase) of each vector are displayed and can be obtained via an interface. The following vectors are available (as applicable for the analyzer configuration), either measured or computed as applicable – Voltage for each phase/channel and between each phase, Current for each phase/channel and Current for Neutral. Additionally, the numerical positive, negative and zero sequence voltage and current amplitudes are available when configured for 3ø4w measurements.
STBY POWER Screen
The STBY POWER Screen enables a user to configure, start/stopand view the results of low power measurements in accordance with EN50564:2011inboth numerical and graphicalformats. The STBY POWER screen can be selected for view from any of the Main Data Screens by pressing the STBY POWER button. The XT2640 Series have the ability to perform low power measurements independently in each VPA, so a single XT2640 can perform up to 3 independent low power measurements. You select which VPA is being configured, controlled and viewed by using the VPA button (the leftmost across the top of the screen). The XT2640 Series perform the low power measurement in accordance with EN50564:2011 paragraph. 5.3.2, which is therecommended method for all low power measurements.
This is only available if option H500 is installable and only if the XT2640 Series are configured for spectrum analysis by the MEAS MODE setting on the MEAS CONFIG screen. Normally the XT2640 Series are used for general power analysis and harmonic analysis purposes, but in some situations they are desirable to perform spectrum analysis on the signals instead of harmonic analysis. If option H500 is installed, then this is possible in the XT2640. An example of this is for measuring distortion in accordance with some avionics specifications which require a spectral analysis with the spectrum analysis being performed with 20Hz resolution for frequencies up to 150kHz (note that this maximum frequency requires W channel types as it is above the 115kHz upper limit for L, S or A channel types).
The XT2640 Series can be configured to measure the spectrum of all signals at the same time to this requirement. As a note, SPECTRUM mode is a single VPA mode; only one VPA can be used in spectrum analysis mode. You do not have to be viewing the SPECTRUM screen for spectrum data to be measured. When configured for spectrum analysis, the XT2640 Series still perform normal power analysis (unchanged) and also harmonic analysis (but limited to the fundamental component). The XT2640 is a single measurement DFT based spectrum analyzer not a ‘scanning’ type spectrum analyzer. When operating in SPECTRUM mode, the HARMONICS screen button shown on the right side of most screens is replaced by a SPECTRUM button. Pressing this button enables the results of the spectrum analysis to be viewed on the SPECTRUM screen.
The XT2640 Series have the capability of data logging up to 16 user defined measurements (or sets of harmonics measurements) at a user defined interval to a file on the front panel USB drive. Each set of data recorded at each data log interval is called a data log record. Data logging is not synchronous to any measurement period, but logs the data at each respective time interval. It is guaranteed that within a single data log record that all data from the same VPA is from the same measurement period. The XT2640 Series use a large internal FIFO buffer to prevent inconsistencies in the write speed of the drive from disrupting data logging. If this buffer is overrun then data logging continues, but some data log records may be lost. This buffer also allows you to initiate data logging before inserting a drive into the front panel USB drive port of the XT2640, but once a drive has been inserted it must remain in place until data logging is stopped.
XiTRON XVIEW UTILITY SOFTWARE
While all XiTRON precision test equipment is designed to be used in a completely stand-alone manner, there are times when external tools can aid or enhance the operation of an instrument. The XiTRON XView software tools and drivers are designed to help easily configure an instrument from a single screen, or are used to view a complete set of measurements in a single screen. Other XView tools are designed for data collection where results can be recorded in an Excel-compatible file for post-processing, insertion into reports, or simply for archival purposes. The XiTRON XView Utility Software comes as standard for XT2640 Series.
|XITRON XT2640 SERIES PRECISION MULTI-CHANNEL POWER ANALYZER
Basic Power Analyzer Chassis, 26S element, 220-260ksps, 22bit effective resolution, 0.1 rdg (V) – 1 Channel
Basic Power Analyzer Chassis, 26S element, 220-260ksps, 22bit effective resolution, 0.1 rdg (V) – 2 Channel
Basic Power Analyzer Chassis, 26S element, 220-260ksps, 22bit effective resolution, 0.1 rdg (V) – 3 Channel
Basic Power Analyzer Chassis, 26S element, 220-260ksps, 22bit effective resolution, 0.1 rdg (V) – 4 Channel