Compliance West 1.2×50-12P 12kV MegaPulse Surge Tester
The Compliance West MegaPulse Surge Tester waveform is in tolerance with the requirements of IEC 60060 from 400V to 12kV and is available in output impedances of 12 or 39 ohms. The IEC 60060 1.2×50 tester is referenced in IEC 60335 and IEC 61730. Built with a relay conforming to the requirements of IEC 60065, Figure 7b. Optionally, the MegaPulse 1.2×50-12 can also perform the tests required in UL 60950 Annex N (in compliance from 2000 to 7000V) for either the 12 or 39 ohm tester.
MegaPulse 1.2×50-12P 12 ohm, 400-12kV, 3 sec charge time, 12 ohm virtual impedance. In accordance with the requirements in IEC 1180-1 Section 6.2; and other Standards by reference. It is acceptable for use with unpowered samples rerquiring a 12 ohm impedance, with no current waveform defined. This tester is in accordance with the measurement requirements of CTL DSH 251B.
The switch is our IER model.
The meter shows the peak of the output waveform into an open circuit. Impedance of the device being tested will attenuate this peak value. Pulse delivery one per minute for pulses above 6kV; two per minute for pulses below 6kV.
Please see Manuals Section below to view waveforms.
This tester can be provided with Option 950-12, which allows testing to IEC 60950 Annex N/IEC 62368-1 Annex D; Line 2.
1.2×50 waveform with 12 Ω or 39 Ω output, 400–12kV
A Compliance West surge Tester. That means it outputs a guaranteed waveform but won’t break the
bank. The tester outputs the waveform it is supposed to, time after time.
Built in the USA. Email and phone support free for life. Reasonable repair and calibration charges. Timely
and helpful customer service always.
Complies with the following standards:
- IEC 60060 Sec 6, Fig 6
- IEC 60335-1 Chap 14 Table 6
- IEC 60065 Annex K, Fig 7b (Optional)
- IEC 61180-1 Sec 6
- IEC 61180-2 Sec 7
- UL 60950 Annex N (Optional)
|Compliance West 1.2×50-12P 12kV MegaPulse Surge Tester|
|Compliance West 1.2×50-12kVP-12
MegaPulse 1.2×50-12P 12 ohm, 400-12kV, 3 sec charge time, 12 ohm virtual impedance
|Compliance West 1.2×50-12kVP-39
MegaPulse 1.2×50-12P 39 ohm, 400-12kV, 3 sec charge time, <42 ohm conventional impedance